WELCOME TO NM3E's RF Wattmeter Pages!
Coaxial Dynamics authorized dealer
Stocking new and used CDI and Bird® items.
COPYRIGHT 2003-2018 NM3E.COM, LLC
(email gets a faster response) My phone # is 610-207-4865>>>email is best
Email questions to email@example.com (NOT my Paypal email)
Every Item is tested and guaranteed NOT DOA = it will arrive Working
CLICK THE LINKS BELOW TO FIND WHAT YOU NEED
If the page looks the same after clicking a link please scroll down
METER RELATED ITEMS:
Prices shown do NOT include shipping.
ALL Content on these pages including Photos are the property of NM3E.com, LLC (copyright 2003-2018)
4410 (OLDER VERSION) 4410A (LATEST VERSION) 4412 (120vac POWERED VERSION)
The meters mentioned all work the same way. By using internal electronics to scale the element reading to the dial setting.
So.....like most other Bird meters the element determines how the meter reads.
Or to put it another way you must use the correct element to get accurate readings for the frequency and power you are measuring.
The element MUST be a 4410 series (4410-#) element or you can ruin the meter.
Many 4410's I buy must be sent for repair / recalibration cause by improper element use.
4410 elements and their common usage are at the bottom of this page.
The meters have a rotary multiplier switch on the front.
The first position is battery check. You need one 9 volt battery that it will check to make sure you get accurate readings.
The Battery is in a small compartment with a swig open door - for easy battery change - NO TOOLS REQUIRED
The next 7 positions are numbered from 100 to 0.1 all of these are % of the maximum power marked on the element.
If an element is inserted that is marked 1000 watts. It will accurately read from 1watt to 1000 watts full scale.
The 100 position reads 100% = 1000 watts full scale.
the 30 position reads 30% of the 1000 = 300 watts full scale
the 10 position reads 10% of the 1000 = 100 watts full scale
the 3 position reads 3% of the 1000 = 30 watts full scale
the 1 position reads 1% of the 1000 = 10 watts full scale
the 0.3 position reads 0.3% of the 1000 = 3 watts full scale
the 0.1 position reads 0.1% of the 1000 = 1 watt full scale
When testing 27MHz 1000 watts or less the 4410-5 element is perfect 25-60MHz 1000watts.
If HF 2-30MHz then a 4410-3 element works up to 1000Watts.
If you run more then 1000 watts but less then 10,000 a 4410 meter with a 10,000w element can be used.
It works the same 100% - 0.1 % but scaled to the 10,000 watt max as follows.
The 100 position reads 100% = 10,000 watts full scale.
the 30 position reads 30% of the 10,000 = 3000 watts full scale
the 10 position reads 10% of the 10,000 = 1000 watts full scale
the 3 position reads 3% of the 10,000 = 300 watts full scale
the 1 position reads 1% of the 10,000 = 100 watts full scale
the 0.3 position reads 0.3% of the 10,000 = 30 watts full scale
the 0.1 position reads 0.1% of the 10,000 = 10 watt full scale
This will require a 4410-4 element 2-30MHz which are more expensive but worth the investment.
If you generate more then 10,000 watts I have high power meter sets here:
ELEMENT NUMBER RATINGS AND COMMON USAGE:
4410-2 10,000 watts 450-2500KHz - old AM radio and induction heating
4410-3 1000 watts 2-30MHz HAM and CB (will work on 160M)
4410-4 10,000watts 2-30MHz HAM and CB (will work on 160M)
4410-5 1000 watts 25-60MHz lower cost 27MHz and 6 meters
4410-6 1000 watts 50-200 MHz - 6 meters, 2 meters, aero mobile
4410-7 1000 watts 144-520MHz 2 meters, 1.25 meters, 70cm, radio navigation
Even thought the 4410 series requires a battery it only reads average / dead key NOT peak / swing.
They cannot be converted to read peak / swing.